All rights reserved. Printed in the Netherlands. All product names are trademarks of their respective companies. Fluke // ScopeMeter. Users Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 . 2 Fluke Corporation ScopeMeter C/B Series, Series and the NEW. Path: /fluke/B/. This is the These manuals are available for download and free of charge. Fluke B/B-C/B-C ScopeMeter Service Manual.
|Published (Last):||1 July 2009|
|PDF File Size:||15.54 Mb|
|ePub File Size:||8.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
All narrow positive pulses shorter than 5 ms are mqnual displayed on the screen. The screen displays the numeric readings of the measurements on the meter input. Table of Contents Add to my manuals Add. Now, you will see a screen like in Figure 15 Figure Note that the Freezing the Screen Peak-Peak reading for input B appears next to the input A You can freeze the screen all readings and waveforms at frequency reading at the top of the screen.
Comments to this Manuals Your Name. Contact your nearest Fluke center for more information. By pressing ‘print’ button you will print only current page.
This function optimizes the position, range, time base, and triggering and assures a stable display of virtually any waveform. To set the trigger delay, do the following: For other countries, a line plug adapter complying with the applicable National Requirements must be used. Printed in the Netherlands. Specified non-tolerance including approvalUL including approval Safety Switching Glitch Detect to Off will suppress the high frequency noise on a waveform.
Fluke 192B User Manual
The horizontal frequency scale is always logarithmic. A marker is To measure rise time, do the following: Tips Using the Tilt Stand Using fluek Tilt Stand Resetting the Test Tool The test tool is equipped with a tilt stand, allowing viewing If you want to reset the test tool to the factory settings, do from an angle while placed on a table.
Chapter 7 Tips About this Chapter Using the Standard Accessories This chapter gives you information and tips on how you The following illustrations show the use of the standard can make the best use of the test tool.
For additional optional accessories, see the ScopeMeter Accessories booklet.
FLUKE B USER MANUAL Pdf Download.
You can choose external triggering with automatic triggering or with edge triggering. Fluke t and t electrical tester datasheet 2 pages. Chapters Table Of Contents DC to scopemster kHz Use the TrendPlot function to plot a graph of Scope or Meter measurements as function of time. Triggering on Scopeemeter Pulses After a reset, the test tool is dc-coupled so that ac and dc voltages appear on the screen. Select the desired trigger slope, and jump to Level: A Select Probe Cal with the arrow message indicates when the DC calibration has completed keys, then accept.
To activate manual ranging, do the following during any Meter measurement: Triggering on Waveforms Using Trigger Delay or Pre-trigger Using Trigger Delay or Pre-trigger You can begin to display the waveform some time before or after the trigger point has been detected. From this position the following: This function can be used to monitor waveforms like motion control signals or the power-on event of an Uninterruptable Power Supply UPS.
Got it, continue to print. Page 7 Contents continued Triggering on Video Signals This example demonstrates how to set up the test tool for Consult the manual that came with your printer to find out printing on a postscript printer with ecopemeter baud rate: Turn the test tool off.
Don’t show me this message again. The test tool has four signal inputs: In this example it is assumed that the Condition.
Enter text from picture: